Hangzhou Lohand Biological Co.,Ltd Add: Zhejiang Hangzhou Jianggan 4F,Bldg 7,No.63,Jiuhuan Rd,Jiubao Town,Jianggan Dist Tel: +86-571-85050999 Mob: +8613123950066 E-mail: email@example.com
There are two types of pH meter installation: flow-through and immersion.
Sewage treatment plants are generally used is immersion installation, such as the sewage treatment plant's pH meter installed in the grit tank outlet overflow tank, the pH here is more representative, and the water flow is stable, the pH meter will not Caused a big impact. Regular maintenance helps to measure the instrument accurately and extend the life of the instrument. Care should be taken that the dedicated cable between the sensor and the transmitter is not damp, otherwise the high impedance, low voltage signal from the electrode will not be transmitted to the transmitter.
If the electrode is not measured, the yellow protective sleeve should be put on, which can make the electrode in a wet state, which is beneficial to prolong the service life of the electrode. Every other month or so, the electrode should be cleaned by first rinsing the attachment with a soft water jet, then immersing the electrode in the cleaning liquid for a period of time, and then rinsing it with water.
The sensor bracket should also be cleaned. After each cleaning, to use buffer solution calibration, the manufacturer usually provides two bottles of standard solution, a bottle of pH equal to 7, used to calibrate the instrument zero; a bottle of pH equal to 4, used to calibrate the signal output slope of the instrument .
The company specializes in producing all kinds of ph meter, welcome to consult procurement.